DIGITEN G1/2" Water Flow Hall Sensor Switch Flow Meter 1-30L/min
Easy to install.
Frequency: F=7.5 * Q (L / Min) error: ± 2% current can not exceed 10mA, 450 output pulses/liters, F=Constant * units of flow (L / min) * time (seconds)
Maximum current:15 mA(DC 5V)
Working voltage range:DC 5-24 V
Load capacity:≤10 mA(DC 5V)
Operating Temp:≤80℃,Operating humidity:35%-90%RH
Allow compression:Water pressure 1.75 Mpa below
Insulation resistance:>100M OHM
Output Waveform:Square Wave,output pulse signal.
G1/2" Flowmeter x 1
We Also Recommend
SENSTREE G1/2" Liquid Display Water Flow Quantitative Controller Meter+Solenoid Valve
G2" inch Water Liquid Controller Display Flow Control Meter Quantitative 10-200LPM
Fácil de colocar!!
Only 2% accurate, but the calibration factor specified as 7.5hz/(l/min)=450pulses/liter seems to be quite off as well. After averaging a couple of tests, I've calculated it to be more like 13.25hz/(l/min) = 796 pulses/liter.
Seems to report relatively accurate flow.
Exactly what I was looking for, and worked perfectly. Just as advertised.
Works as Required